Worldwide indirect to direct on-demand eye doctor support refraction system via a remote administration tool mobile application on any portable Electronic device with broadband wireless cellular network technology 4G ,5G , 6G or Wifi wireless network protocols to interconnect both systems.

ABSTRACT

The present disclosure describes clinical workflows, methods, and systems used to perform an indirect to direct subjective refraction to a patient with a mobile smartphone application that works as a encrypted remote administration tool in any portable electronic device and interconnect both systems via by 4G, 5G, 6G, or Wifi. According to various embodiments, an eye doctor may utilize a remote administration tool (RAT) or (RAS) remote access software application on a portable electronic device (PED) (smartphone, tablet, or laptop) to view and control the main control base (MCB) anywhere in the world to interconnect both systems. The eye doctor can perform an on-demand live subjective vision refraction via RAT technology. Furthermore, the eye doctor can control the (MCB) that can control, exam chair, digital phoropter, vision chart software, robotic phoropter arm, exam chair height, exam room lights, and near robotic chart arm anywhere in the world.

RELATED APPLICATIONS

This U.S. Patent Application is a continuation of U.S. Provisional Application No. 62/982,040 filed Feb. 26, 2020 titled “A indirect to Direct on demand eye doctor support Refraction system via Remote administration tool using any portable Digital Device by 4G, 5G or Wifi to interconnect both systems”.

BRIEF DESCRIPTIONS OF THE DRAWINGS

FIG. A Illustrate a pictorial diagram of how the connection from the eye doctor (PED) and the main control base in the exam room may occur.

FIG. B illustrates a flow diagram of a method for performing interconnection from the eye doctor (controller) to the main control base (MCB) to provide a indirect to direct subjective refraction using an mobile application that works as a Remote administration tool to a patient and sending a eye glass prescription to patient or location.

FIG. C Illustrates embodiments of list of patient data and the office geolocation displayed during a starting interconnection of both systems.

FIG. 1 Illustrates embodiment of what may be displayed on the eMCB and what can be controlled during (RAT) interconnection (robotic phoropter and Vision chart laptop or display).

FIGS. 1A-1B Illustrate embodiment of a request to accept or decline connection service from the on demand eye doctor to Main control base from Assistant or patient in the exam room.

FIG. 1C—Illustration embodiment of what may be displayed on a MBC and on the patient smartphone after the scan QR code on the MCB.

FIG. 1D—Illustration embodiment of what exam room with robotic remote equipment operating and a the display patients PED and the (controllers PED).

FIG. 1E—Illustrates a flow chart of a method for a Remote or non remote assistant via (RAT) to guide a patient into the exam room and provide instructions while also helping with the eye doctors interconnection to the system.

FIG. 2—Illustrates embodiment of what may be displayed in a mobile clinic or non mobile eye exam clinic and what is viewable by the controller via RAT app using 4G, 5G, 6G or wifi.

FIGS. 3A-3B Illustrates embodiment of how the Robotic digital phoropter may be controlled by the main control unit (wire cable or bluetooth depending on system used).

FIGS. 4A-4B Illustrates embodiment of Second setup where the Assistant is inside the room and the patients smartphone connection with eye doctor can be miracast to a laptop or display via wired HDMI, bluetooth or wifi.

FIG. 4B—above

FIGS. 5A-5B Illustrates embodiment method and display of the eye doctor sends patient prescription via his PED to the patient.

FIG. 5 Illustrates embodiment display of the eye doctors PED of the signed prescription via finger to touch screen

FIG. 6 Illustrates embodiment of a mobile Vision exam room unit that can be controlled remotely by an eye doctor PED, or assistant controller PED to perform remote administration tool controlled subjective refractions services via wireless 4G, 5G or 6G generation broadband cellular network technology.

DETAILED DESCRIPTION

Many health practitioner facilities in the world are incorporating telemedicine technologies in their establishments. Some practitioners use telemedicine video/audio devices to connect with remote patients or other doctors to provide medical service too distant hospitals. This has become a new normal after the covid19 pandemic. For example, a remote radiologist may provide his medical service in a relatively large hospital in a city. This same radiologist may provide telemedicine video advice, viewing, consultation or assistance to patients and/or other medical professionals in a secondary hospital located in a remote area. Some of these telemedicine encrypted video consultation services are provided by new companies that provide low-cost HIPPA, HIPAA, PIPEDA, and GDPR compliant video conferences to doctors and patients by just scanning link URL with no need to download software.

Telemedicine systems are becoming normalized worldwide. However, the use of remote telemedicine robotics control is still in its infancy. Amazingly such robotic devices like the da (Vinci Surgical System) is a robotic surgical system made by the (American company Intuitive Surgical). It is designed to help surgery using a minimally invasive approach controlled by a surgeon from a console. The (da Vinci System) consists of a surgeon's console typically in the same room as the patient and a patient-side cart with three to four interactive robotic arms managed from the console. Just as these new robotic medical devices emerge the use of new innovative ways to provide medical service will emerge.

The present disclosure provides various methods, system methods, apparatus, and clinical workflows in telemedicine robotics to operate a robotic digital phoropter, exam equipment and room equipment in a mobile or non mobile exam room via a (end to end encrypted HIPAA compliant) remote administration tool technology. The following is for utilizing the main control base (MCB), a computer running on any operating system and that may run multiple software and multi emulators and a remote administration tool to improve connection control and view to a controller (eye doctor or assistant). Interconnecting the office MCB with the (controller) to provide an indirect to direct subjective refractions to a patient via and mobile application that works as or with a Remote administration tool. In some situations the system can switch to a service sharing that may not provide direct service. Instead of providing the primary service, the system will instead provide a secondary eye doctor's connection to connect and perform an indirect to direct subjective refraction remotely. The new users or (controller) set up a personal profile with a name, phone number, state license number other information, and payment preference, which could be a credit card, e-commerce payment system or, in some cases, cryptocurrency address. This payment will be used to pay the in-depended contracted eye doctor for the service provided world wide.

Service is generally accessed via mobile app. The application may ask to contain Users (controller) set up a personal profile with a name, phone number, state license number other information, and payment preference, which could be a credit card, e-commerce payment system or, in some cases, cryptocurrency address, or cash. The remote administration tool mobile application can be of commercial or non-commercial use. Remote administration tool application may be used on any portable electronic device any broadband cellular network technology 4G, 5G, 6G or Wifi to interconnect on both systems. The robotic digital phoropter (FIG. 1D 352), and exam room equipment (FIG. 1D FIG. 1 FIG. 4B) can all be controlled remotely by a controller that controls the main control base MCB. As used herein, a telemedicine video interconnectivity via personal smartphones between the eye doctor controller and patient in the eye exam room is established via both the controller and patient's personal or portable electronic device PED. (FIG. 1D 350 d,353 p). The connection can also occur via live chat; you can easily communicate with a patient via text and/or screen sharing via any cellular network technology or wifi.

A secured Doctor to patient indirect to direct subjective refraction peer to peer support service. This support service system allows subjective refined refraction to be made via two mobile applications that work as remote administration tools (RAT) or remote access software (RAS) (Figure A—38, 42, 42,40,39) without the eye doctor having to be in the Exam room. The remote administration tool (RAT) software and/or Remote access software (RAS) can be of commercial or non commercial means. The eye doctor can be on-site in different rooms or off-site anywhere in the world. The patient can be anywhere in the world in a mobile or non-mobile exam room.

An example of an office inside a mobile clinic or non-mobile clinic location is the following pictorial representation. (FIG. 6) (FIG. 4B—24).The exam room's main control base can be connected to an online medical appointment booking service if needed. However, it will also have the possibility of using online medical appointment booking service 3rd party commercial or non-commercial with a patient PED app. The exam room equipment door and lights can also be controlled via the RAT via the MCB. The support connection is made by (RAT or RAS) software or application via 4G,5G, 6G wireless network or wifi (FIG. 3A—11) (FIG. 3B—10,1.).

The robotic digital phoropter and vision chart (DP&VC) can be controlled directly by the MCB by an assistant in the office (FIG A—43,44) via bluetooth or wired connection (FIG. 3B—5,6) depending on the system used. The MCB may contain a universal adapter that facilitates the control and connection too many digital phoropter in the market to control the digital phoropter.

FIG. 1E illustrates examples embodiments of a situation where an assistant can not be physically present during the service. Where the virtual assistant can control the main control unit to talk to the patient, open the door remotely and virtually instruct the patient to clean the equipment with alcohol wipes and sit in the exam chair for the beginning of the session. The virtual assistant will be able to tell the patient to put his or her glasses in an automatic lens-meter that can read the glasses on a push of a button. Automatically the lensometry reading of old glasses will be uploaded to the main control base software.

The virtual assistant can also tell the patient to situate him or herself in front of a camera that works like as an auto-refractor and/or keratometer both eye will be scanned. The virtual assistant can also tell the patient to situate him or herself in-front of a secondary camera controlled remotely that works to find the patient pupillary distance. The results will automatically be sent to the main control base. The virtual assistant will then give access to the selected eye doctor (username and password) to control the main control base. The virtual assistant will then tell the patient to scan Qr code (FIG. 1C, FIG. 1D 310,311) on the main control base for the patient to connect directly with the eye doctor via the patient's PED RAT mobile application.

During this time, the eye doctor will connect to the main control base and have full control of the exam room and Exam equipment via his PED. The main control base (MCB) and universal adapter (FIG. 3B—2) can control the digital phoropter (FIG. 3B—4) & digital vision chart (DP&VC) (FIG. 3B—3,9) via software in the MCB and software or mobile application that uses a remote assistant tool (RAT) running in the background (FIG. 3B—1). The main control base can be of any operating system Android, windows, IOS or Linux. (FIG A—43) and can run an emulator with software or software to control a digital phoropter and digital vision acuity chart dependent or independent (FIG. 3B—8,25,26,7,3, 9). The eye doctor can have any portable electronic device (PED) (FIG. 4A—19,12) (FIG A—42) with any operating system Android, Windows, IOS or Linux, with an installed (RAT) software or (APP) (FIG A—42,41).

The main control base has to have both the (RAT) Application (FIG. 1B—27,31,33,32) and phoropter operating software running (FIG. 2B—33,27) at the same time to be controlled by a controller (eye doctor) (FIG. 3A—11) (FIG. 3B—10,1) anywhere in the world. The support controller (eye doctor) (FIG. 1A) needs to have the (RAT) software (APP) open (FIG. 1A—28). The support controller (eye doctor needs to have the (RAT) (APP) running to receive a request to control the main control base (MCB) by the controller portable electronic device (FIG. 1A—30,29,28,) (FIG. 1B—32,31,27,33). The main purpose of the MCB is direct control of the digital phoropter and Vision Acuity Chart (DP&VAC) via wired or Bluetooth connection (FIG. 3B—2,5,6,4,3,9,7,8,25,26). The main control base can also connect to the internet via Wifi, 4G or 5G connectivity (FIG. 3B—10) (FIG. 3A—11).

The (RAT) Remote Assistant tool mobile application or software runs inside the existing operating system of the (MCB) (FIG. 4B—18,14) (FIG. 2B—27, 33). The controller can be anyone with an Electronic portable device (FIG. A—42) (FIG. 3A—11) with 4G,5G, 6G or wifi connectivity from anywhere in the world with a cellphone signal or wifi. A portable electronic device (PED) such as a smartphone, Tablet or Laptop can control the main control base (MCB) via 4G,5G, 6G or wifi (FIG A—41,38,40,39,43). The MCB can be remote-controlled over Internet protocol networks using an applet running on the communication device (FIG. 2B—27,32,33), and the support controller (eye doctor) can control the (MCB) by using a (RAT) application running on his or her portable electronic device (smartphone, tablet, or Laptop) (FIG A—42) When the technician, optician or Doctor (FIG. 1B—33,27,31,32 (A)) sends a support request to an on-demand (eye doctor) (FIG. 1A—29,27,30,28, (DR) he or she can accept or decline the request for (RAT) interconnection.

If and when the (eye doctor) becomes a (controller) and accepts the request, he or she has to insert a user name and password (FIG. 1A—28) for the office requesting the support they wish to provide the support service. Each interconnecting session is encrypted, And can have two-factor authentication for security if needed. The (RAT or RAS) remotes administration/Remote access software program installed within its system randomly generates a user name and password authentication for the (eye doctor) controller to be able to log in and connect to the MCB (FIG. 1B—33 & FIG. 1A—28). This Secures the peer-to-peer network (FIG A—39,38,41,40) to control the (MCB) (FIG A—39,43), which directly controls the Digital phoropter control or digital phoropter software and Vision Chart (FIG A—44,43).

This random user name and password facilitate a secure connection to any (eye doctor) who decides to accept (FIG A, FIG B—201,200) to become a (support controller) to provide service with their portable electronic device. The controller is indirectly controlling the Digital phoropter (FIG. A—44) as the connection is made from the eye doctor controller (DPED) (FIG. 3A—11 & FIG. 3B—2,1,10) to the (MCB). By way of indirect to direct connection, the eye doctor (DPED) can control the (MCB) which will directly control the robotic digital phoropter & Vision Chart (DP&VC) (FIG. 3B—4,3,2,5,6) and all equipment in the exam room. The (RAT or RAS) can use commercial software or non-commercial service to achieve its servers to middle-man to do the connection (FIG. A—38) and connection set-up, but most of the time, the relationship between client and server will be PEER TO PEER (FIG. A—39) (Controllers DPED to MCB).

The controller (eye doctor) can indirectly remotely operate the robotic digital phoropter near chart, digital phoropter & vision chart, to refine a prescription (FIG. 2A—30,28,36,34,35,37) from a patient behind the digital phoropter (FIG. 4B—20,15,16,17). The patient eyes can be tested for best visual acuity facilitating a proper prescription vision check (FIG. 3B—3,9,4). A patient prescription can be refined from past existing prescription glasses data obtained in the office (FIG. 3B—25,).

In a situation where the Assistant can be in the exam room. An assistant in the exam room will read or import patient auto-refractor/keratometry readings result for both right and left eye (FIG. 4B—18,18A) (FIG. 2A—36,37) but the system can also import to MCB automatically or via the eye doctor MCB control. The assistant will also read existing glasses prescription by lensometry of both right and left eye (FIG. 2A—37). The eye doctor will average out the lensometry reading and the auto-refractor reading to acquire a starting point prescription in each eye individually for both right and left eye (FIG. 2A—36,37). This will be the patient's primary base refraction inputted into the digital phoropter (FIG. 3B—8,5,6,4) before remote prescription subjective refraction can be obtained by an eye doctor on on-demand connection via smartphone connection at bluetooth or long-distance (wifi, 4G, 5G or 6G (FIG. A—41,40)) (FIG. 3A—11) (FIG. 4A—19,13,21,22,23,12) from anywhere in the world.

Again the controller (eye doctor) will be connected to the (MCB) via the (RAT). The on-demand Optometrist or Ophthalmologist (eye doctor) anywhere in the world can perform the indirect (RAT) Subjective refraction (IRATSR) and can communicate with the patient via smartphone weblink. The secured video/audio, text or audio connection will be HIPAA compliant encrypted video & voice or voice call (FIG. 4B—272,274,271,275,20) (FIG. 4A—270,19,13,12). For privacy, the patient can also have the option of using wireless over-the-ear headphones (FIG. 4B—271) with a microphone (FIG. 4B—275) to establish a clear voice connection with the eye doctor (FIG. 4A—270,19,13,12). The support controller (eye doctor) can also use portable wireless headphones with a microphone to maintain clear communication with the patient (FIG. 4A—19,270). In some cases interactive smartphone button selection system can be used by the patient to remotely select the lenses choices.

The Optometrist or Ophthalmologist (eye doctor) will ask the patient which lenses seem better refining sphere, cylinder and axis in both eyes when viewing the smallest line he or she can see on the vision chart (FIG. 3B—3,9). The eye doctor will perform monocular subjective vision checks via the vision chart. This to each eye and make the necessary adjustment to the prescription if needed. The doctor will verify patients' best Visual acuities with monocular (one eye at a time) (FIG. 3B—9,3), binocular vision testing (both eyes together) and binocular near vision to all patients connected. A last-minute monocular pinhole vision test using the robotic digital phoropter on each eye for distance will validate if the prescription. (FIG. 4B—15,20,275,271). This will be help refinement before releasing an eye glass prescription to the patient. The goal is to achieves the best-corrected visual acuity with the best refine prescription.

The eye doctor can validate the patient visual acuities (VA) and make the necessary correction to the refractive prescription. This will be done to acquire the patient's best-corrected vision acuity in each eye (FIG. 5A—254,251,262) (FIG. 5B—,253,265,250,261). The eye doctor (controller) can view the chart displayed screen mirroring to his PED via the RAT application. (FIG. 5A—251) (FIG. 5B—260) (FIG. 5B—250,263,264). A complete indirect (RAT) Subjective refraction can be performed and a new eye glasses prescription may be generated after the refined subjective vision acuity check. (FIG. 5B—253) (FIG. 5—300). After the refined indirect (RAT) Subjective refraction is done the eye doctor can send out the patients prescription via screenshot picture, printing from the main control base, or email.

SUMMARY OF INVENTION

In part, workflows, apparatus, method and system of establishing interconnection via a mobile application that works as a remote administration tool (RAT) to control and view the main control base to control all robotic medical equipment and room equipment and providing an indirect to direct on-demand eye doctor support subjective refraction service with any portable electronic device via 4G, 5G, 6G wireless cellphone communication or wifi to a patient in an exam room. The exam room can be a mobile clinic room or a non-mobile clinic. The rooms can be anywhere globally with wireless communication capabilities via broadband cellular network technology 4G, 5G, 6G, or wifi.

This is not limited to accepting a first patient in an exam chair in a room that can be remotely controlled. The remote control can be accessed by an eye doctor or assistant at or away from the location. The room contains remote locks and doors that can be locked or open at the location or remotely. The room includes an exam chair that can be controlled remotely or not remotely. The room contains a robotic digital phoropter that can be controlled remotely or not remotely. The room includes a mechanical lever that controls the near-reading chart of the phoropter. The room consists of a robotic phoropter arm that can be positioned in front of the patient face safely. The room contains lights that may be remotely controlled. The room includes a vision chart that can be a display or laptop running a vision chart software. The vision chart can be remotely controlled. The room contains the ability to have broadband cellular network technologies 4G, 5G, 6G, or wifi, depending on location. The control of the exam room equipment is possible by software running the main control base.

The main control base can run secure password-protected electronic medical record (EMR) software, which can also be controlled. All data from auto-refractor readings and lensometry readings can also be viewed inside the main control base. Only one patient data is viewed at a time in each session. All of this equipment can be controlled by the main control base and therefore can be reached by any (remote controller) eye doctor or assistant. The assistant can be a (remote controller) if they are not able to make it to the exam location that day. The (controller) can be an assistant or eye doctor. The controller can be anywhere in the world. The exam rooms and the patient can be anywhere in the world also. The (remote controller) can establish a connection to the main control base via a remote administration tool that interconnects both systems via broadband cellular network technology 4G, 5G, 6G, or wifi, depending on equipment. The controller can also ask the patient to be an assistant if needed in some cases since the main control unit is at arm's length from the patient. An eye doctor can control what can be viewed on the main control base by the patient.

Upon entering the room, the patient can view a QR code on the main control base. The patient can scan the Quick Response code (QR code) with his smartphone camera app that will direct the patient to an URL or application. The scan will establish a direct link to a web-based telemedicine communication that designates a secure video/audio or audio communication between the eye doctor or (controller) and patient. After the patient establishes video/audio or audio communication via smartphone, the patient and eye doctor will begin the session. The patient may also be able to use button (1 or 2) realtime interactive system to communicate choices to doctor virtually. The eye doctor will answer any questions and ask the patient acuity vision choices before and during the subjective refraction and vision chart. The eye doctor will conduct a subjective refraction to the patient. During the session, the eye doctor can lower the phoropter arm remotely. The eye doctor asks the patient to position his or her eyes behind the digital phoropter lenses. The doctor will then provide a live refined subjective refraction session. The eye doctor will ask the patient questions to determine distance lens choices and near vision lens choices to acquire the best eyeglass prescription for both the patients right and left eye. Following the refined subjective refraction completion, he can sign the eyeglass prescription and print it at the facility or email. 

What is claimed:
 1. a workflow, method, and system to provide remote administration tool controlled subjective refractions services to patients worldwide via mobile applications interconnecting both systems via broadband cellular networks technology 4G, 5G, 6G or wifi wireless internet connection comprising: achieving a secure connection from the controllers personal electronic device mobile application to the main control base via a end to end encrypted (HIPPAA) Health Insurance Portability and Accountability Act compliant (remote administration tool) (RAT) in a mobile clinic or non mobile clinic; displaying multiple software controls systems on dashboard running on the main control base that can be shared with controller via RAT screen share and remote control; Where the mobile clinic can be any vehicle an ambulance, a bus, a train, a boat, an oil rig, or airplane. opening list of patients scheduled with the geolocation in the world of the exam room on the main control base and on the eye doctors (controller) portable electronic device application; controlling the main control base to remotely open the door to let patient in and give the patient instructions virtually or not virtually via the main control base; direct the main control base to ask patient to use his or her smartphone to scan unique QR code link URL to establish virtual eye doctor HIPPAA compliant encrypted audio/video, and/or text connection with eye doctor with the ability to provide screen sharing and control to patient smartphone; controlling the main control base remotely with patients vision pre-testing and HIPAA encrypted electronic medical record data via HIPAA encrypted screen sharing and the RAT running on PED and MCB; controlling the main control base remotely to control all software running and multi-emulated screens running that controls: door locks, digital phoropter control, exam chair, robotic phoropter near chart, robotic phoropter arm, lens-meter software, Auto-refractor control, exam room lights and vision chart software and other system in office; viewing via screen share control or automatic transfer to MCB the automatic autonomous lens-meter, and auto-refractor results on the main control base via emulators and/or not via emulators; controlling and viewing the main control base interface remotely to control the and adjust the exam chair vertical height and move the phoropter arm to position the a person to start the subjective refraction session; controlling and viewing the main control base interface remotely. Controlling the in-depended digital phoropter control software and vision chart software that controls the digital automatic phoropter and Vision chart. acquiring the best corrected refine patients distance and near subjective eyeglasses prescription with a Remote administration tool screen share and control software via broadband cellular networks technology 4G, 5G, 6G or wifi wireless internet on both systems; controlling and viewing the main control base to virtually sign a patients finalized eyeglass prescription and printing on the main control base and/or emailed the glasses prescription to patient smartphone anywhere in the world; directing, upon conclusion the indirect to direct remote subjective refraction session, the phoropter arm retracts, the exam chair lowers and the eye doctor may answer any questions from the patient and give patient instruction to clean and leave the room via the smartphone HIPPAA compliant encrypted video or audio and/or text connection. after the first patient disconnects, the patient may leave the room; subsequently, the next patients may enter the room. the first patient that was already refracted may be followed up for the second part of the eye health examination by an optometrist or ophthalmologist on site with a live eye doctor, or telemedicine doctor in any location or same location if not have been seen already on the same day or different day by the same or different doctor.
 2. The method of claim 1, wherein: initiating a smartphone app, tablet app or laptop app that serves as a remote administration tool providing screen share and remote control to an eye doctors to a main control base via broadband cellular network technology 4G, 5G, 6G or wifi. The controller is further configured to receive two or more wireless communication protocols if needed. receiving a secure interconnection from the controller (eye doctor or assistant) via a commercial or non commercial remote administration tool running inside the application in the eye doctor portable electronic device and the main control base. Verifying that all multiple software are running on the main control base via multi-emulators if needed. verifying that all software including the software running on emulators can be control via HIPPAA compliant encrypted remote administration tool connections.
 3. The method of claim 1, comprising: viewing a list of patients scheduled and the geolocation of the exam room on the application running on the main control unit anywhere in the world. viewing the list and selected patient from the eye doctors (controller) personal electronic device via the HIPPAA compliant encrypted application remote administration screen share and control.
 4. The method of claim 3, further comprising: receiving data from the (virtual or live assistant) access of the selected patient from the patient list via the main control base interface via HIPPAA compliant encrypted RAT screen share or email. verifying that the assistant can also share with the eye doctor patient list on online medical appointment booking service for medical providers if needed. Verifying that all software running inside the main control base or a software running inside an emulator inside the main control unit are gathering data and sending data.
 5. The method of claim 1, further comprising: initiating PED to MCB software connection to unlock and open the exam room door and via text to speech ask patient to walk in, sit down in exam chair in front of the main control unit and/or follow the instruction written words on the main control base. the main control base may also speak prerecorded instructions to the patient. if the assistant is at the remote exam room they can help with the instruction also.
 6. The method of claim 5, further comprising: initiating the eye doctor connection and control to the main control base to direct the main control base via visual text or voice prerecorded text to speech to ask patient to use his or her smartphone to scan unique QR code and/or link URL to establish virtual eye doctor audio/video, or audio and/or text connection. The link URL connection will be via the eye doctor and patient smartphone and via secondary screen sharing tool if necessary. verifying that the link also opens an (1 or 2 selection mobile application) to be used by a patient so he or she can manually select lenses via two buttons (one or two) choice system. Each number will correspond to a lens during the exam and the eye doctor give control to the patient of the digital phoropter if needed also. Instructing the patient on how it is used to choice selection lens in subjective refraction.
 7. The method of claim 1, wherein: initiating the eye doctor or virtual assistant control and view connection to the main control base remotely and view and control secure password encrypted patients data (prior or same day) pre-testing and patient electronic medical record data via screen sharing by a commercial or non commercial remote administration tool and/or with a computer with a multi-emulator installed running software on the main control base and that can be controlled by the smartphone application that runs remote administration tool in the background.
 8. The method of claim 1, wherein: controlling the vision chart by the main control base software. Controlling the MCB vision chart software via RAT screen share and remote control with the controller personal electronic device RAT mobile application. Vision chart can also work as a color vision testing device to remotely change the color plates to gather color vision testing. all connection from he MCB softwares to the Vision chart display may be via broadband cellular network technology, infrared, bluetooth or wifi depending on the system used and can be controlled from anywhere in the world.
 9. The method of claim 1, wherein: controlling the main control unit MCB to control and screen view the MCB interface to remotely control the software that controls the robotic medical or non medical equipment in the exam room. controlling the MCB that controls the digital phoropter to perform distance and near subjective refraction. all from the same control base and its display via remote administration tool via screen share and remote control that runs inside a smartphone application on the eye doctors personal electronic device using broadband cellular network technology 4G, 5G, 6G or wifi to interconnect both systems.
 10. The method of claim 1, wherein: advising patients on remote positioning and adjusting the patient in the exam chair by remotely controlling the exam chair software. adjust the proper vertical height and adjusting the phoropter arm to position as to set the patient at eye level with the vision chart and the digital phoropter behind in position to patient eyes using RAT via screen share and remote control to the equipment control software.
 11. The method of claim 1, wherein: controlling MCB via RAT screen share and remote control manipulate the office robotic equipment and lights from the location navigating the digital phoropter control and Vision chart control to start subjective vision refraction session.
 12. The method of claim 1, wherein: remotely transmitting the automatic lens-meter readings, and auto-refractor results on the main control base via remote administration tool screen share and control or automatic transfer bluetooth, or emulators with screen share capabilities. an emulator is hardware or software that enables one computer system to behave like another computer system.
 13. The method of claim 12, wherein: using the patient data from the automatic lensometry reading from patients right and left eye old glasses prescription. where the information was gathered from the patient placing his or her eye glasses on a specific automatic semi-autonomous lens-meter machine where the machine can read the glasses prescription by the patient automatically or by the press of a button where results are automatically sent to the MCB or received via RAT screen share or by emulator screen sharing. an emulator is hardware or software that enables one computer system to behave like another computer system.
 14. The method of claim 13, wherein: gathering and recording the auto-refractor results from past technician evaluations or same-day remote auto-refractor results. both readings are automatically sent on the main control base via an application via, bluetooth, wired or wifi, depending on the system used or via RAT via screen share and remote control and emulators screen sharing. Averaging both eyes automatic lensometry readings and auto-refractor readings to get a tentative starting point subjective refraction prescription lenses on the digital phoropter for the right and left eye lenses. Inputing the information o the MCB so as to lenses automatically change to the averaged prescription reading calculated.
 15. The method of claim 1, wherein: subsequently, to adjusting the patient properly and remotely situating the robotic phoropter arm, transmitting the tentative prescription to the digital phoropter and precisely adjusted the patient behind the automated phoropter the eye doctor can begin the session. starting the indirect to direct subjective refraction. refining each eyes lens prescription individually. Start refining the sphere, the cylinder and the axis of each eye. during the refining asking patient to select the buttons that correspond to the lens via the (1 or 2) application, verbal or visual sign language. asking the patient to choose between dual options for lenses with the patient smartphone application if needed. Viewing results via screen smartphone screen or text. Viewing the results of the patients selection via app with secondary device answers 1 or
 2. changing lenses according to selected answer. the patient may use this when taking the indirect to direct subjective refractions to send response to eye doctor when asked in wish lens does the letter look better. the patient can also have the option of doing a verbal choosing of lens one vs lens two via RAT screen share and remote control with broadband cellular network technology 4G, 5G, 6G or wifi wireless internet connection.
 16. The method of claim 3, wherein: subsequent the eye doctor will refine the prescription of the right and left eye independently for the sphere power, cylinder power, and axis via RAT screen share and remote control. the eye doctor may perform a red/green chromatic aberration test. the eye doctor may perform binocular balance test via the phoropter controlling the lenses on the phoropter. after this he will Perform monocular vision test and binocular test to acquire the best acuity for patients final prescription. anytime during the exam the the eye doctor may perform monocular lens sun burst dial or clock dial test via the vision chart to verify astigmatic correction. after this a pinhole test is performed to find optimal VA. also patient may control the digital phoropter lens change for subjective comparison of lenses if needed via two remote control (1 & 2) buttons on the chair or portable electronic device app and under eye doctor video/audio and/or audio supervision and instruction in some cases. in the case of Video the patient can use his or her finger 1 or two fingers to chose the lenses.
 17. The method of claim 1, wherein: subsequent the eye doctor starts a near subjective refraction, a near robotic chart moves down so that near vision chart is visible to the patient. accommodative test can be performed via remote means to acquire appropriate add power for presbyopic patient. the near chart then retracts when the testing and add power is found by the doctor and this is done via an application on the main control base using bluetooth, wire or wifi depending on the system used.
 18. The method of claim 9, further comprising: receiving an indication that the eye doctor (controller) wishes to conclude the (RAT) subjective refraction session and directing the phoropter arm to retract automatically or via control away from the patient face. also that the exam chair moves down position automatically or via remotely control by the controller via RAT screen share and remote control.
 19. The method of claim 1, where: delegating view and control patients electronic medical record data via a RAT screen sharing & control. where the first eye doctor delegate secondary connection to a secondary eye doctor. where the secondary doctor can view and control the main control base software installed and running in the main control unit. where the eye doctor can view and control and explain the patients electronic medical record data, retinal health images, external ocular health pictures of all the ocular parts all from his or her portable electronic device and running via a secure connection with a commercial or non commercial remote administration tool software on both devices from any location in the world all via broadband cellular network technology 4G, 5G, 6G or wifi wireless internet connection.
 20. The method of claim 1, further comprising: that this system via a mobile app can be switched to provide or delegating of service to other medical doctors and/or eye doctors via secure encrypted connection. that the system can also determine the fees and terms on which independent eye doctors providers may provide HIPPAA compliant encrypted remote administration tool refractions services to patients via the mobile app or software. mobile application users (controller) may set up a personal profile with a name, phone number, state license number other information, and payment preference, which could be a credit card, e-commerce payment system or, in some cases, cryptocurrency address. 